This instrument is designed and made as per standards T 0615-2011 and ASTM D721 Wax Content Test for Bitumen by distillation in JTG E20-2011 Specification and Test Methods of Bitumen and Bituminous Mixture for Highway Engineering and SH/T0425 Standard Test Method for Wax Content Test of Bitumen. It is used to determine the wax content in the petroleum asphalts by cracking distillation.
Main technical features
This instrument adopts method of distillation cooling to determine the wax content of sample. The instrument has two parts: Upper part contains stirring motor, temperature sensor, air flow distributor and hook for sample vessel. Lower part is the cooling filter.
The special designed air flow distributor and hook for sample vessel make the instrument more reasonable and convenient for use.
The constant temperature bath adopts glass material. It has good heat preservation. There is no mist on the observation window. Convenient to observe.
This instrument equips a high accuracy digital temperature controller with clear temperature show and simple temperature set.
This instrument is floor-type structure. It can determine 3 samples at the same time. The test efficiency is high.
Main technical parameters
1. Power supply： AC（220±10%）V，50Hz
2. Heating power： 700W
3. Cooling power： 1033W(-20℃, )
4. Cooling mode：Compressor refrigeration
5. Test temp.：-20℃
7. Working environment：15℃～28℃，
9. Total power consumption：≤1800W
1.Vacuum drying oven：Model DZF-6020（Recommend）
2. Burning furnace： Model SYD-0615-1 Cracking furnace（Recommend）
3. Balance： 0.1mg,0.1g one set for each
No. 1, Jianlong N. Rd., Jiulongpo District,Jiulong Industrial Park, Chongqing, China