The instrument is designed and made as per the regulation of cracking distillation stove in the petrochemical industrial standard of People's Republic of China SH/T 0425-2003 and ASTM D6307 “Test Method for Wax content of the Asphalt ” and T0615-2011 “Asphalt Wax Content Test (Distillation Method)” in the industrial standard of People’s Republic of China JTG E20-2011 “Standard Test Methods of Bitumen and Bituminous Mixture for Highway Engineering ” .It is the specified equipment and matched with the “SYD-0615 Asphalt Wax Content Tester”,It also satisfies the needs of petroleum products .
Main technical features
This instrument adopts the mode of electric heating ,and the constant temperature can be controlled the range of 550℃+10℃.The heating power can be adjusted continuously and it can satisfy the needs of the distillation of samples .
The instrument adopts heating furnace and elevator-platform.Two parts are independently and in use two is made one ,it is convenient to use .
The housing washer of heating furnace adopts humanized design of heat insulation and reduce the effect of the heat radiation to the operator .
The temperature controller adopts LCD display, single chip control technology. Advanced technology and small overshoot .It can reach to 550℃+10℃constant temperature ,which only needs 15 minutes at normal atmospheric temperature.
Main technical specifications
1. Power supply: AC(220±10%)V, 50 Hz
2. The electric furnace heating power: 2000W
3. Heating control: solid adjustable pressure model can be adjusted continuously.
4. Constant temperature control: 550℃+10℃
5. Ambient temperature: ≤35℃
6. Relative humidity: ≤85%
7. Total power consumption:≤2100W
8. Dimension: heating furnace: 220mm*330mm*400mm
9. Weight: 5kg
No. 1, Jianlong N. Rd., Jiulongpo District,Jiulong Industrial Park, Chongqing, China